What is the sensitivity of
Tau cameras in Watts per m^2 ?
The reported Noise Equivalent Temperature Difference (NEdT)
for a Tau 320 camera is <50mK at f/1.0. This can be translated
to change in power per unit area to yield that the minimum resolvable
change in flux is <0.14 watts / sq meter at f/1.0.